It is a flow computer that is ideally suited to applications requiring remote monitoring, gas flow calculations, data archival, supervisory setpoint control, and closed-loop control.
Keys allow numerical entries.
W.4.Mounting hole diameter.4 mm (0.37.The die-cast enclosure protects the electronics from physical damage and harsh environments.Memory Reset: When used during power-up, Reset switch initializes communication port hardware and communications port processing for all ports.MVS interface Type: High-speed, multi-drop, serial interface with power for as many as four MVS units located up to 1220 m (4000 ft) from the FloBoss 407 crack remote administrator 2 2 unit.Options The FloBoss 407 supports the following options: Remote Multi-Variable Sensor (MVS) unit.All other marks are property of their respective owners.Storage Temperature: 50 to 85C (58 to 185F).Eight-terminal connector provided on I/O board.FloBoss 407 Flow Manager.Optional MVS (integral or remote).
Data: 512 KB battery-backed sram.
Pipestand Mounting: Mounts on 2-inch pipe with U-bolt mounting kit (included).
Any type and combination of I/O modules can be used.The MVS inputs and I/O modules are configured in the same manner as the built-in inputs.Emerson Process Management Remote Automation Solutions Marshalltown, IA 50158.S.A.Two built-in process inputs are provided for interfacing to measurement instrumentation.Download, report, transcript 2:FB407 Specification Sheet April 2007 Remote Automation Solutions Website: m/Remote D301072X012 FloBoss 407 Flow Manager The FloBoss 407 Flow Manager provides the functions required for measuring, monitoring, and managing flow on single or multiple meter runs.Watchdog Timer: Hardware monitor expires after.2 seconds and resets the processor.Print, print page 1, rename the bookmark, delete from my manuals?H.3.The MVS sensor is a high-accuracy device that can be integrally or remotely mounted.EIA-422/EIA-485 (RS-422/RS-485) for asynchronous serial communications.
Approvals Standard Version: Approved by CSA for hazardous locations Class I, Division 2, Groups A, B, C, and D, T4, and.
The I/O consists of two built-in process inputs, a Multi-Variable Sensor (MVS) port, and accommodations for up to four plug-in I/O modules, any of which can be a hart Interface Module.
Discrete Input (source or isolated).